EUROPEAN SOUTHERN OBSERVATORY
KeywordsTime Reference System
Time Signal Distribution
Accuracy 10 MIcroseconds
Fiber Optic Transmission
MetadataShow full item record
RightsCopyright © International Foundation for Telemetering
Collection InformationProceedings from the International Telemetering Conference are made available by the International Foundation for Telemetering and the University of Arizona Libraries. Visit http://www.telemetry.org/index.php/contact-us if you have questions about items in this collection.
AbstractThe necessity of supplying precise time information in large telemetry ground stations and astronomical observatories is very similar. Therefore the way of solving this problem as it is done in the Very Large Telescope of the European Southern Observatory can be easily adopted to telemetry stations and ranges, especially when fiber optics are used. The European Southern Observatory (ESO) is building a new observatory in Chile for the Very Large Telescope (VLT). This VLT consists of 4 telescopes, each of them has a primary mirror diameter of 8 meters. the control architecture is based on workstations and VMEbus computers. The VMEbus computers are distributed over the whole building and are using real time operating system. Since the availability of the Global Positioning System (GPS) the generation of highly accurate timing signals on remote locations without the use of expensive Cesium standards does not create problems any more. However, distribution of a timing signal to many computer with high accuracy is an issue. The accuracy of the commonly used IRIG B-code is not adequate if the requirements are in the 10 microseconds range. This paper presents the design of a timing system that is adopted to the VLT. An overview of the requirements of the Time Reference System (TRS) is given. These requirements have been defined on the basis of experiences with the timing system of the ESO NTT telescope. The hardware units are described. These are a Central Time Standard, a Time Distribution System and a VME Time Interface Module. The distribution is based on fiber optic transmission, using a simple digital modulation that outperforms the analog IRIG B modulation. The Time Interface Module in the computer does not only perform the timing signal decoding but contains also user-programmable timers that are synchronously clocked from the time source. Presently all units of the TRS have been tested and the series production of the distribution and the Time Interface Modules are in progress.
SponsorsInternational Foundation for Telemetering